KP-1 is a SWR and Power meter employing current transformer covereing the frequency range of 1.8MHz through 30MHz.
■ A large scale analogue cross meter and 7 segment LED allows for easy reading.
■ Meter has a built-in white LED back lighting allowing for readouts in low level light.
■ Meter has four ranges to measure power from 10W to 10KW. Digital display is not affected when changing measuring range.?
■ Digital display indicates SWR value.
■ PEP power readings with adjustable hold time, while operating SSB/CW.
■ Over-load protection circuit protects your linear amplifier and antennas. User adjustable trigger value of SWR(1.4 ～ 2.5) activates the protection circuit when SWR
exceeds the threshold level. Protection circuit opens the TX GND line between exciter and linear amplifier.
|KP-1 Product specification|
|Frequency range||1.8～30MHz||Insertion loss||Leaa than 0.1 dB|
|Power measurement range||0～10KW||Input-and-output impedance||50Ω|
|Electric power range||FED 500W /1KW / 5KW / 10KW
REF 125W / 250W / 1250W / 2500W
|Input-and-output||M type （SO-239）|
（A projection thing is not included.）
|Power measurement accuracy||±７％（Full-scale value ）||Weight||Approx1.7Kg|
|SWR measurement minimum electric power||Approx 8W||Accessories||Operation manual
|① Cross Meter
You can read FWD/ REF power and SWR withthis analog meter. The crossing point
of needlesshows the SWR value.② Digital Display
You can select functions with the buttons below the display. When the function is selected, the button shines blue.
③ FWD Button
④ REF Button
⑤ SWR Button
⑥ PEP Button
|⑦ DEL Button
Digital display shows the power going into your antenna FWD‐(minus)REF when this button is pressed.⑧ PROTECT Lamp
When the SWR becomes higher than the value you set up, it is activated. When activated, the connection between 12 TX GND and 13 RELAY IN is disconnected inside of KP‐1.
⑨ POWER Switch
⑩ Range Select
|⑪ Power jack
For DC12V adaptor plug.⑫ TX GND Terminal
To be connected with TRX TX GND.
⑬ RELAY Terminal
|⑭ RF OUT Connector
SO‐239 to connect your antenna⑮ RF IN Connector
SO‐239 to connect with your TRX
A Russian MAI-SSTV event is planned from the International Space Station for Wednesday, June 5 from 12:00-16:00 GMT and June 6 from 11:30-15:30 GMT. ARISS say they expect transmissions to be at 145.800 MHz FM in SSTV mode PD120. Based on th... Read more
True kit – soldering required (about 40 hours to build boards, assemble, cal and test) Modular (start small, add modules as your budget allows) Wide selection of IF filters, plus the ability to slide the 2nd and 3rd IF filters back an... Read more
See More: http://www.amateurlogic.com Read more
Amplifier, HT, VHF, Solid State, 144-148 MHz, 3/50 W In, 200/200 W Out, UHF Fem, SO-239, 13.8 Vdc, 25 A, Each Add this amp to your 2 Meter HTor mobile... Read more
Description： The Baofeng BF-F9+TP is the upgraded version of BF-F9+ with the trip power 1w/4w/8W/. With its 128 channel capacity and up to 12 hours of... Read more